click image to enlarge
72" CR/DR Lateral Cervical Spine
Model 400 - $595
A two week complimentary trial is available. Include the name of the x-ray machine when ordering (model number is not needed).
Computerized imaging limitations are when one part of the anatomy requires 5X or more overexposure to image an adjacent
area whose thickness requires this large increase to bring the anatomy to the proper density. This is true in imaging the
lateral cervical spine when imaging the thicker area of C7/T1. The filter is designed to reduce the need for the swimmers
view by increasing visualization of the C7/T1 in the lateral cervical spine view.
The filter balances the exposure between
the mid-cervical spine area and the C7/T1 junction. In the upright position the top of the shoulders will normally fall
slightly below the level of the C7/T1 junction on most people. It is the thickness of the trapezius muscle at the level of
C7/T1 that diminishes viewing density of the C7/T1 junction. The reduced thickness of the wedge on the distal portion
increases the density by reducing absorption which results in an increase penetration of the C7/T1 junction.
Using the Filter
The technologist slides the filter in the collimator rails until it stops: opens the hinged filter, collimates, closes the filter and exposes.
The starting technique is labeled on the filter.
Lateral Cervical Spine Features:
- The swimmers view has viewing limitations due to the patients overlapping anatomy. Positioning can be difficult.
The reader would prefer Interpreting an image of C7/T1 in a true lateral position without the imposition of anatomy
- The mr measured with the phantom in place: Entrance dosage mid level cervical spine without filter 69.2 mr 20 mas 80 kVp, Mid level cervical spine with filter 40.9 mr 95 kVp 63 mas T1 area with filter 96.4 mr
The swimmers view would be about 5 X the mr.
- The filter can be used in cervical spine surgery and ER backboard patients. There is a base size available for portables.
Please call for more informaton (877) 429-9329
Slides easily onto the collimator rails
Image from Merrill's Atlas Volume one 12th edition page 53 Compensating filters